Verifying the precision of a measurement procedure against a performance claim and estimating the bias (CLSI EP15-A3)
Video tutorial, with step-by-step instructions and example files.
Text version and example files
Watch on YouTube
Learn how to verify that a measurement procedure meets performance requirements before introducing a new measurement procedure into the laboratory, after maintenance to the measuring system, or upon failing a proficiency test or inspection. In this tutorial you will use the CLSI EP15-A3 procedure to verify the precision against the manufacturer’s claim and estimate the bias using proficiency testing materials.
This video applies to Analyse-it version 5.50 and later.