We've been developing intuitive high-quality statistical software at an affordable price, backed up by fast friendly customer service for over 25 years...
Office hours: 9am-5pm GMT Mon-Fri
It's Tuesday 5:02 PM and the office is closed - e-mail us and we'll get back to you as soon as possible.
Estimate the precision of the measurement procedure at multiple sites and samples.
The worksheet opens showing 4 columns. The Sample column identifies the levels of the analyte in the sample (6 levels). Laboratory identifies the site (3 laboratories), Run identifies the run (5 runs per site). CA19-9 (kU/L) identifies the measured value for 5 replicates of sample in each run.
The analysis report opens.
The variability plots show a simple visual assessment of the closeness of agreement between the measured quantity values. The purple lines show the mean of each run, the light blue lines show the mean of each laboratory, and the dark blue line the overall grand mean.
You should observe the scatter of the points to ensure there are no obvious problems. No individual measurements stand out as highly aberrant relative to the bulk of the data and none of the plots exhibit any apparent drift capable of distorting the results.
The abbreviated variance components table shows the required precision statistics expressed numerically as the standard deviation (SD) and coefficient of variation (CV).
The detailed variance components table show the precision expressed numerically as the chosen measure of imprecision along with a confidence interval for each component.